Organization:Apollo 14 Passive Seismic Experiment
Close-up view⋅of the PSE on the Moon's surface
|Uses||To detect vibrations and tilting of the lunar surface and measure changes in gravity|
The Apollo 14 Passive Seismic Experiment (PSE) was placed on the lunar surface on February 5, 1971, as part of the Apollo 14 ALSEP package. The PSE was designed to detect vibrations and tilting of the lunar surface and measure changes in gravity at the instrument location. The vibrations are due to internal seismic sources (moonquakes) and external (meteoroids and impacts from the spent S-IVB and LM ascent stages). The primary objective of the experiment was to use these data to determine the internal structure, physical state, and tectonic activity of the Moon. The secondary objectives were to determine the number and mass of meteoroids that strike the Moon and record tidal deformations of the lunar surface.
The PSE unit was constructed principally of beryllium and had a mass of 11.5 kg, including the electronics module and thermal insulation. It was housed in a drum-shaped enclosure 23 cm in diameter and 29 cm in height. The enclosure was rounded on the bottom and rested on a leveling stool. The PSE consisted of two main subsystems, a sensor unit and an electronics module. The sensor unit contained three matched long-period (LP) seismometers aligned orthogonally in a triaxial set to measure one vertical and two horizontal components of surface motion. The horizontal component seismometers were very sensitive to tilt and were leveled to high accuracy by means of a two-axis motor-driven gimbal operated by ground command. A third motor adjusted the vertical component seismometer in the vertical direction. A fourth, short-period (SP) seismometer with a resonant period of 1 second measured vertical motion at a peak sensitivity of 8 Hz and a response range from 0.05 to 20 Hz. A thermal shroud and 6-W heater for thermal control comprised the rest of the experiment package. The thermal shroud was aluminized mylar which covered the instrument and the ground surrounding the base out to about 75 cm radially. A gnomon and level sensor were mounted on the top center of the shroud. The shroud was modified slightly from the one used on Apollo 12. Total power drain varied from 4.3 to 7.4 W.
The seismometers consisted of an inertial mass on a sensor boom suspended by springs and hinges, a capacitor plate and a damping magnet. The LP seismometers could function in a flat-response mode and in a peaked response mode. In the flat response mode, the LP seismometers had a natural period of 15 s. In the peaked-response mode, they acted as underdamped pendulums with a natural period of 2.2 s. Sensitivity to ground motion peaked sharply at 0.45 Hz in peaked response mode with a useful frequency range of 0.004 to 2 Hz. Maximum sensitivity was enhanced by a factor of 6 in the peaked response mode, but sensitivity to low-frequency signals was reduced. All seismometers could detect ground motions as small as 0.3 nm. At tidal frequencies, gravitational acceleration was measured by monitoring the feedback current used to center the seismometer mass. The sensitivity of the instruments was 0.008 mgal. The lunar surface impacts of the spent S-IVB and LM ascent stages were used as external calibration sources for the seismometers. The known mass and velocity of these stages at surface impact and the lunar impact point coordinates enabled the computation of energy generated at impact and the point of energy application. (The calibration characteristics were determined by measuring seismometer response to these energy sources.)
The seismometers were deployed on February 5, 1971, and operated at reduced gain while the astronauts were on the lunar surface and turned to maximum sensitivity for most of the time after that. The ALSEP central station was located at. The passive seismic experiment was deployed 3 meters north of the central station. At deployment, the LP vertical seismometer was unstable in the flat response mode, so all three LP seismometers were operated in peaked-response mode until 17 November 1976 when the problem was rectified. The gimbal motor operating the leveling for the LP horizontal Y seismometer showed intermittent malfunction. The Apollo 14 ALSEP suffered several periods of loss of signal, in one case for over two months. All data were lost for these periods and commanding of the instruments was not possible. Seismic disturbances were noted throughout the lunar day, but particularly near sunrise and sunset, these were believed to be due to expansion and contraction of the mylar shroud and/or the cable to the central station. The instrument was configured to standby mode on 30 September 1977 as part of the ALSEP station shutdown.
- "Passive Seismic Experiment (PSE)". NASA. https://nssdc.gsfc.nasa.gov/nmc/experiment/display.action?id=1971-008C-04.
- Description of Apollo Seismic Experiments R. Yamada
- Apollo 14 Seismic experiments Lunar and Planetary Institute
Original source: https://en.wikipedia.org/wiki/Apollo 14 Passive Seismic Experiment. Read more